BRAND offers the ideal pipette for every hand: BRAND’s new Transferpette® S. Transferpette® S models are are the perfect manual pipettes for demand-. Transferpette® electronic. F I R S T C L A S S · B R A N D. Gebrauchsanleitung. Operating Manual. Mode d’emploi. Instrucciones de manejo. 3. -8/ . Testing Instructions (SOP) testing of the piston operated pipette Transferpette®. BRAND additionally provides a calibration service (see chapter 7).
|Published (Last):||17 March 2006|
|PDF File Size:||5.91 Mb|
|ePub File Size:||4.16 Mb|
|Price:||Free* [*Free Regsitration Required]|
Parts will differ slightly depending on nominal volume of instrument. Hold the instrument vertically, with the sample in the tip for approximately 10 s. In order for the liquid to reach its end position, leave the tip immersed for a few seconds.
Consult the manufacturer for manhal in temperatures outside of this range. Pipetting c Let the pipetting button slide back slowly. Inspect the piston and O-Ring-seal for contamination. Safety Instructions Safety Instructions Please read the following carefully!
Safety Instructions Please read the following carefully! Also See for Transferpette S Operating manual – pages.
Checking the Volume Depending on use, we recommend inspection of the instrument every 3 to 12 months. Fitting the tip Use the correct tip according to the volume range or the color code.
Don’t have an account? Don’t show me this message again. Limitations Of Use Limitations of Use Purpose The pipette is an air-displacement system for pipetting aqueous solutions with medium density and low to medium viscosity.
Laboratory Equipment Brand dispensette Operating Manual pages. Optimum analysis tgansferpette can only be obtained with quality tips. Ensure that the tip is securely seated.
Page 8 Pipetting c Let the pipetting button slide back slowly. To print the manual completely, please, download it.
BRAND Transferpette S Operating Manual
Autoclave the complete pipette without any further disassembling. Discharge sample a Place the pipette tip against the wall of the vessel.
To do this, affix a tip, and aspirate a sample. This instrument may sometimes be used with hazardous materials, opera- tions, and equipment. Page 9 Pipetting Ejecting the tip Hold the pipette shaft over a suitable disposal container and press the tip ejection key to the stop. Warranty Information Warranty We shall not be liable for the consequences of improper handling, use, servicing, operating or unauthorized repairs of the instrument or the consequences of normal wear and tear especially of wearing parts such as pistons, seals, brajd and the breakage of glass as well as the failure to follow the instructions of the operating manual.
It is the responsibility of the user of this pipette to consult and establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
BRAND TRANSFERPETTE S OPERATING MANUAL Pdf Download.
Inspect the pipette tip cone for damage. By pressing ‘print’ button you will print only current page. The form can also be downloaded from www. Pipetting 5 ml and 10 ml instruments should only be used with the PE filter installed see page Pipetting Ejecting the tip Hold the pipette shaft over a suitable disposal container and press the tip ejection key to the stop.
It is beyond the scope of this manual to address all of the potential safety risks associated with its use in such applications. Adjustment — Easy Calibration The instrument is permanently adjusted for aqueous solutions. If the pipette operation is clearly inaccurate, or if the instrument must be adjusted for solutions of different densities and viscosities or specially- shaped pipette tips, adjustments can be made using the Easy Calibration Technique.
Pipette tips are disposables items! Comments to this Manuals Your Name. ISO prescribes rinsing the pipette tip once with the sample liquid prior to the actual pipetting process.